|Year : 2017 | Volume
| Issue : 4 | Page : 114-117
Measurement of central corneal thickness by different techniques
Reem Hassan Ibrahim Azzam, Manal Ali Kasem, Abeer Khattab, Hossam Mohamed Ali El-Fallal
Department of Ophthalmology, Faculty of Medicine, Mansoura University, Mansoura, Egypt
|Date of Submission||08-Jun-2017|
|Date of Acceptance||08-Oct-2017|
|Date of Web Publication||8-Mar-2018|
Manal Ali Kasem
Department of Ophthalmology, Faculty of Medicine, Mansoura University, Gomhoria Street, Mansoura, 35511
Source of Support: None, Conflict of Interest: None
Purpose The aim of this study was to compare the central corneal thickness (CCT) measurements by three different techniques: pentacam, noncontact specular microscopy, and ultrasound biomicroscopy (UBM) devices.
Patients and methods This prospective observational cross-sectional study was performed on patients seeking medical care at the outpatient clinic of Ophthalmology Center at Mansoura University. They were consecutively recruited and selected, and measurement of CCT by pentacam, UBM, and specular microscopy was done. Thereafter, the results were compared as an evaluation study of all techniques.
Results This study included 200 eyes of 100 patients. The age of the included patients in the study ranged from 20 to 50 years. The mean age was 34.56±9.94 years. The CCT ranged from 465 to 630 µm, with a mean of 532.88±34.15 µm when measured by pentacam. When measured by specular microscopy, it ranged from 456 to 601 µm, with a mean 531.92±33.64 µm, and when measured by UBM, it ranged from 463 to 602 µm, with a mean 533.97±31.10 µm, with P value of 0.824 (nonsignificant).
Conclusion CCT measurements obtained by UBM tend to be thicker than those obtained by pentacam and noncontact specular microscopy, but the measurements of all three are strongly positively correlated with each other. So, any of these devices can be easily substituted by the other for the measurement of CCT.
Keywords: corneal thickness, pentacam, specular microscopy, UBM
|How to cite this article:|
Azzam RI, Kasem MA, Khattab A, El-Fallal HM. Measurement of central corneal thickness by different techniques. J Egypt Ophthalmol Soc 2017;110:114-7
|How to cite this URL:|
Azzam RI, Kasem MA, Khattab A, El-Fallal HM. Measurement of central corneal thickness by different techniques. J Egypt Ophthalmol Soc [serial online] 2017 [cited 2019 May 23];110:114-7. Available from: http://www.jeos.eg.net/text.asp?2017/110/4/114/226940
| Introduction|| |
Measurement of central corneal thickness (CCT) has a great value in different fields of ophthalmology and optometry, especially in diagnosis and treatment of corneal disorders ,.
CCT is also an important parameter in refractive surgery. It helps in predicting the long-term complications such as postsurgical keratectasia, which shows the importance of accurate corneal thickness measurement .
Different equipments are used now to measure CCT such as ultrasound pachymetry, contact and noncontact specular microscopy (NCSM), optical coherence tomography, pentacam, ultrasound biomicroscopy (UBM), partial coherence interferometry, and confocal microscopy ,,,,,,.
The value of CCT measurements may differ among these instruments. This is based on a variety of techniques and devices, and each has its own advantages and disadvantages .
The aim of this study was to compare the CCT measurements by three different techniques: pentacam, NCSM, and UBM devices.
| Patients and methods|| |
This is an observational cross-sectional study, and it was performed on patients seeking medical care at the Outpatient Clinic of Ophthalmology Center at Mansoura University. They were consecutively recruited and selected, with the aim of measuring CCT by pentacam, UBM, and specular microscopy and then comparing the results as an evaluation study of all techniques.
Patients included in this study aged from 20 to 50 years, were of both sexes, and had spherical equivalent ranged from +2 to −2 D with healthy cornea and intraocular pressure less than 21 mmHg.
Patients excluded in this study had the following criteria:
- Corneal diseases such as corneal dystrophies, degeneration, ectatic diseases, and keratitis.
- Systemic diseases affecting eyes such as diabetes mellitus and Herpes Zoster.
- Past history of ocular surgery such as refractive and cataract surgeries.
- Contact lens wearers.
In this study, 200 eyes of 100 patients of both sexes between 20 and 50 years were enrolled after obtaining complete informed consent from all of them in the period from January 2014 to December 2014 and approval of the Institutional Review Board of Mansoura Faculty of Medicine.
The candidates underwent comprehensive eye examinations, which included the following:
- Uncorrected and best-corrected visual acuity.
- Slit lamp biomicroscopy examination (Haag-Streit, Bern, Switzerland) for detection of any corneal abnormalities and ensuring clarity of anterior chamber.
- Measurement of intraocular pressure was performed using the Goldmann applanation tonometer attached to Haag-Streit slit lamp.
- Fundus examination using 90 D Volk noncontact lens.
Measurements of CCT were done between 9 a.m. and 2 p.m. at least 2 h after waking up. Examination started first by pentacam (Tomey Corporation, Nagoya, Japan). After getting pentacam readings, CCT was assessed by NCSM (Tomey Corporation) followed by UBM (Humphrey, Fresno, California, USA).
The data analyses were performed using SPSS program version 15.00 (SPSS Inc., Chicago, Illinois, USA). Qualitative data were presented as numbers and percentage. Quantitative data were described by mean±SD and range. The results of CCT measurements obtained using each device were compared by using a repeated measures analysis of variance test. Pearson’s correlation coefficient was used to illustrate strength of relation between two variables. P value less than 0.05 was considered significant statistically.
| Results|| |
The study included 200 eyes of 100 patients who underwent CCT measurement using pentacam, specular microscopy, and UBM. The age of the included patients in the study ranged from 20 to 50 years. The mean age was 34.56±9.94 years.
Central corneal thickness measurements
The CCT ranged from 465 to 630 µm, with a mean 532.88±34.15 µm, when measured by pentacam. When measured by specular microscopy, it ranged from 456 to 601 µm, with a mean 531.92±33.64 µm, and when measured by UBM, it ranged from 463 to 602 µm, with a mean 533.97±31.10 µm, with P value of 0.824.
Correlation between pentacam, specular microscopy, and UBM
There was a strong positive correlation between specular microscopy and pentacam (r=0.949, P>0.01; Pearson’s correlation coefficient). This correlation was more than that between pentacam and UBM (r=0.901 probability, P>0.01; Pearson’s correlation coefficient) and between UBM and specular microscopy (r=0.894 probability, P=0.01; Pearson’s correlation coefficient), as shown in [Table 1].
|Table 1 Correlation among pentacam, specular microscopy, and ultrasound biomicroscopy|
Click here to view
The highest positive correlation of CCT readings was by pentacam and specular microscopy ([Figure 1]a), followed by the correlation between pentacam and UBM ([Figure 1]b) and between UBM and specular microscopy ([Figure 1]c).
|Figure 1 (a) A scattered plot of central corneal thickness measurements by specular microscopy versus pentacam. (b) A scattered plot of central corneal thickness measurements by ultrasound biomicroscopy versus specular microscopy. (c) A scattered plot of central corneal thickness measurements by ultrasound biomicroscopy versus pentacam.|
Click here to view
| Discussion|| |
In this era of developing refractive surgeries, accurate estimation of CCT becomes mandatory, so several techniques are available now for estimation of CCT. Each of them has its own advantages and disadvantages. Several research studies have investigated the comparison of CCT measurement between different devices to determine whether any one of these devices can be more accurate than others .
In this study, CCT measurements by pentacam, specular microscopy, and UBM were estimated in eyes with healthy corneas, where a good agreement among the three methods was found. This study mentions the agreement of these three techniques with each other in a single research.
In this study, CCT measured by pentacam ranged from 465 to 630 µm, with a mean of 532.88±34.15 µm. Mean CCT measured by specular microscopy was 456–601 µm, with a mean of 531.92±33.64 µm, and when measured by UBM, it ranged from 463 to 602 µm, with a mean of 533.97±31.10 µm.
Comparison of pentacam with noncontact specular microscopy
The results of this study showed that the mean CCT measurements by pentacam were thicker than those measured by specular microscopy by 0.96 µm. Ucakhan and colleagues, found that CCT measurement by NCSM was thinner than that obtained by pentacam, and this agrees with this study ,,,,.
Comparison of ultrasound biomicroscopy with pentacam
In this study, the mean CCT measurements by UBM was thicker than the mean measurements by pentacam by 1.09 µm. Kanellopoulos and Asimellis , found that the UBM had a mean CCT of 545.1±28.3 µm, which was more than that given by pentacam (538.0±30.7 µm), and this agrees with this study’s results.
This may be explained by corneal hydration following the application of local anesthesia.
Comparison of ultrasound biomicroscopy with noncontact specular microscopy
The mean CCT measurements by UBM were thicker than the mean measurements by specular microscopy by 2.05 µm. It can be explained by the topical anesthesia that was used before UBM, which may alter the corneal thickness, the location where the ultrasonic probe is applied.
On the contrary, Tam and Rootman  found that the mean CCT by specular microscopy was 572 µ which was significantly greater than that measured by UBM (555 µ). This is explained by the compression of the epithelium produced by ultrasound probe. This produces a slightly thinner CCT when measured with UBM.
| Conclusion|| |
CCT measurements obtained by UBM tend to be thicker than those obtained by pentacam and NCSM, but measurements of all three are strongly positively correlated with each other. So, any of these devices can be easily substituted by the other for measurement of CCT.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Patwardhan AA, Khan M, Mollan SP, Haigh P. The importance of central corneal thickness measurements and decision making in general ophthalmology clinics: a masked observational study. BMC Ophthalmol 2008; 8:1.
Sng C, Ang M, Barton K. Central corneal thickness in glaucoma. Curr Opin Ophthalmol 2017; 28:120–126.
Malecaze F, Coullet J, Calvas P, Fournie P, Ame JL, Brodaty C. Corneal ectasia after PRK for low myopia. Ophthalmology 2003; 110:267–275.
Suzuki S, Oshika T, Oki K. Corneal thickness measuremesnts: scanning-slit corneal topography and noncontact specular microscopy versus ultrasonic pachymetry. J Cataract Refract Surg 2003; 29:1313–1318.
Bovelle R, Kaufman SC, Thompson HW, Hamano H. Corneal thickness measurements with the Topcon Sp-2000P specular microscope and an ultrasound pachymeter. Arch Ophthalmol 1999; 117:868–870.
Bechmann M, Thiel MJ, Neubauer AS. Central corneal thickness measurement with a retinal optical coherence tomography device versus standard ultrasonic pachymetry. Cornea 2001; 20:50–55.
Yekta A, Hashemi H, KhabazKhoob M, Dostdar A, Mehravaran S, Heravian J, Fotouhi A. Comparison of central corneal thickness measurements with Pentacam, Orbscan II, and Ultrasound Pachymeter. Iran J Ophthalmol 2009; 21:51–57.
Tam ES, Rootman DS. Comparison of central corneal thickness measurements by specular microscopy, ultrasound pachymetry, and ultrasound biomicroscopy. J Cataract Refract Surg 2003; 29:179–1184.
Sacu S, Findl O, Buehl W, Kiss B, Gleiss A, Drexler W. Optical biometry of the anterior eye segment: interexaminer and intraexaminer reliability of AC-Master. J Cataract Refract Surg 2005; 31:2334–2339.
Kaufman SC, Musch DC, Belin MW. Confocal microscopy: a report by the American Academy of Ophthalmology. Ophthalmology 2004; 111:396–406.
Lackner B, Schmidinger G, Pieh S, Funovics MA, Skorpik C. Repeatability and reproducibility of central corneal thickness measurement with Pentacam, Orbscan, and Ultrasound. Optom Vis Sci 2005; 82:892–899.
Ucakhan OO, Fujioka M, Amano S, Ciolino JB. Recent advances in ophthalmic anterior segment imaging. Br J Ophthalmol 2006; 91:551–557.
Fujioka M, Nakamura M, Tatsumi Y. Comparison of Pentacam Scheimpflug camera with ultrasound pachymetry and noncontact specular microscopy in measuring central corneal thickness. Curr Eye Res 2007; 32:89–94.
Al-Ageel S, Al-Muammar AM. Comparison of central corneal thickness measurements by Pentacam, specular microscopy, ultrasound pachymetry in normal and post-LASIK eyes. Saudi J Ophthalmol 2009; 23:181–187.
Tai LY, Khaw KW, Subrayan V. Central corneal thickness measurements with different imaging devices and ultrasound pachymetry. Cornea 2013; 32:766–771.
De Bernardo MD, Borrelli M, Mariniello M, Lanza M, Rosa N. Pentacam vs SP3000P specular microscopy in measuring corneal thickness. Cont Lens Anterior Eye 2015; 38:21–27.
Kanellopoulos AJ, Asimellis G. Forme fruste keratoconus imaging and validation via novel multi-spot reflection topography. Case Rep Ophthalmol 2013; 4:199–209.